What is the capacitance metering range ?
1132396
http://www.fluke.com/fluke/inen/digital-multimeters/fluke-115.htm?PID=559931000nf to > 1000 micro f.
1132397
From 1 nano Farrah to 9999 micro Farrah according to specs, I only used for electrolytic capacitors and for smaller capacitors I am using a Honeytek A6013L, that is not expensive and do the work very nicely.
1501384
It depends on what you mean by "good". It will measure the low frequency or DC capacitance and you can decide if the value is within what you consider the proper spec. However, it will not tell you if the capacitor can withstand the the rated voltage. It also will not tell you the leakage rate for the capacitor. To determine the leakage rate you will have to charge it to a specific voltage, measure the voltage before disconnecting it from the voltage source, and then measure it again some time later, using a voltmeter with a very high impedance front end, so that the act of measuring the voltage won't discharge the capacitor. It also won't tell you the capacitance as a function of frequency. I hope that this helps you.
2034343
There is a tolerance in its output. I do not remember what it is but the instructions that came with the unit and its maker can give you the specifics.
1689301
Look at the values for the capacitance listed on the device and compare the two.
822499
Rated for up to 6oo volts AC or DC. (Any location). I should note that my initial interest in this product was peaked by thinking the description suggested that it was capable of reading and measuring voltages by mere inductance, which turned out not to be the case. It still uses the probe tips for voltage readings, and using inductance for voltage detection like any other proximity sensor (light and beep) indicator.
896188
I used this on 60Hz, 220V Line to Line. I assume the measurements are correct though, never really tested accuracy at this specs.
398889
Yes, but they are very basic. You have to know capacitance to interpret the readings correctly.
2058016
The instructions with my tester were very basic, but accurate. Written instructions included with any cheap capacitor tester, or multimeter are minimal at best. If you don't know how to use a cap tester it's best not to use one, until you take a class or two, large caps can kill you if you don't know what you're doing.
2247353
Yes, for all practical purposes. Incredible, inexpensive device accurately measures R, L and C with high precision. Caps test provides capacitance and ESR (equivalent series resistance) simultaneously, in seconds. There may some electrolytics in high voltage applications that would require leakage tests under the working voltage but these aren't seen too often. In a test, a 220 uf electroyic 63 wv, with a 500 ohm shunt resistor added, still gave an accurate read of capacitance and ESR. This replaces an ohmmeter, cap meter and ESR meter. I am buying another one. Hope this info is helpful.
1978242
A standard 2.4 Ghz RF device has a range of 10 meters or about 33ft. This range can be affected by electrical interference or line of site loss, but those are issues you can account for during setup. To be comfortable I'd suggest being no more than 25' from the USB receiver.
1285852
This is a great meter, I have purchased two and gave one to my father. The answer to your question is yes, on page 26 which is 4.2.7 you will see that the range runs from 4nF through 20uF and resolution of 1pF through 100 nF.Chris
794132
Hello,This is a great analog meter which is perfect for testing all different types of diodes but not a great tool for testing capacitors. I would highly recommend a ESR meter for testing Capacitors. Hope this helps.
1333877
I wouldn't do it if I was you. You can use a cap with a higher voltage rating but not a higher capacitance rating.
1894190
It's not stated due to the flux capacitor reading being negligible at best. I would suggest staying at a Holiday Inn Express!
2034340
I haven't actually measured it, but I have to assume the output voltage is very stable when the input voltage is within the specs of the device ... that's the whole point of the device.
2247352
Don't know but probably not reliably. It would depend on the amplitude of the signals the tester uses to execute the tests and the nature of the circuit that the cap is in. There are lots of comments/blogs out there about this tester if you google around.
2169103
This meter is designed for appliance service technicians, and is not a laboratory instrument. In practice you will see two places past the decimal when working with 12 volt systems and one place past the decimal when working with 120 volt systems. (It is a 4 digit display.) According to the manual 400mV range has 0.1mV resolution ± (0.5% + 4 digits) That only works if you are measuring something below 400mV, otherwise you get an out of range error. Read more here: http://www.kleintools.com/sites/kleintools/files/instructions/CL2000_TrueRms-Digital-Clamp-Meter-Instructions.pdf